Local Storage seems to be disabled in your browser.
For the best experience on our site, be sure to turn on Local Storage in your browser.
Glass Body pH Half-Cell Electrode with BNC Connector HI2111B
The HI2111B is a glass body pH half-cell electrode with a BNC connector. This sensing electrode is made with high temperature glass and is to be used in conjunction with a reference half-cell electrode. This design consideration is ideal for use in strong alkaline solutions as well as for general purpose measurements.
- High Temperature (HT) Glass
- Half-Cell Design
- BNC Connection
|Note||Heated transport required during winter months.|
|Range||pH: 0 to 14|
|Recommended Operating Temperature||0 to 100°C (32 to 212°F)|
|Tip / Shape||spheric (dia: 9.5 mm)|
|Recommended Use||general purpose, strong alkaline solutions|
Hanna Instruments offers a wide variety of pH electrodes that are designed for many different applications. The type of glass used, body material, type of junction, type of reference and electrolyte used are just some of the design considerations.
The HI2111B uses high temperature (HT) glass, spherical bulb, and glass body.
High-Temperature Glass Formulation
The measurement of pH at very high temperatures is detrimental to the sensitive glass bulb and will shorten the life of it. A pH electrode with general-purpose (GP) glass will have a resistance of 100 megaohms at 25°C while the resistance of HT glass is around 400 megaohms at 25°C. As the HI2111B is used at elevated temperatures the resistance decreases to approach the that of GP glass. The HI2111B is suitable to use with samples that measure from 0 to 100°C.
The spherical bulb is for general purpose use. Other tip shapes include conic for penetration and flat tip for surface measurements.
The glass body is ideal for laboratory use. The glass is resistant to many harsh chemicals and is easily cleaned. The glass body also allows for a fast transfer of heat to the internal reference electrolyte. The mV generated by the reference cell is temperature-dependent. The faster the electrode reaches equilibrium, the steadier the reference potential.
The HI2111B uses a BNC connector. This type of connector is universal in that it can be used on any pH meter that has the female BNC probe input. Other types of connectors include DIN, screw type, T-type, and 3.5mm to name a few. These types of connectors tend to be proprietary for a particular type of meter and are not interchangeable.